VS-MI-100-12-AlmaSoprano-B-808nm Alma 12bars laser stack blue side plate
Product Code: VS-MI-100-12-AlmaSoprano-B-808nm
Availability: In Stock
808 Alma Soprano Repair Description
This VS-MI-100-12-AlmaSoprano-B is a 12-bar laser stack with a blue side plate. The electrodes are on the back of the stack.
It is used for Alma Soprano handle piece repairing and replacement.
Oriental-laser focused on diode laser bar assembling for more than 16 years and we have seen various alma soprano diode laser stacks. Also, we can use the same Jenoptik bar as Alma used to recondition the power of the laser soprano power.
12bars Alma soprano handle Repairing procedure
>Choice A: Send the old broken handpiece to our factory. We will do the process of checking, repairing, cleaning and etc. Then send the fresh new handpiece back to you.
>Choice B: if you have basic assembly capacity, you can send the diode laser stack only to our factory. We remove the old bar and mount the new bar to recover the laser power of the stack.
>Choice C: Measure your stack and confirm the water pipe hole and screw hole position. Then press the button below to buy a ready-made stack that matches your old handpiece.
808 Alma Soprano repair Advantages
>Save Money. Replace the broken diode stack only. Low cost and convenient
>Long lifetime. we use diode laser bars from Germany. The newest Austin mounting technology will be applied when repairing. After repair. The handpiece will be a new one. lifetime is much longer than before
>All technical support available. We majored in diode laser mounting, software, and spare parts for over 16 years. All the technologies about laser hair removal machines are open to you.
808 Alma soprano laser stack Specification
Model No VS-MI-100-12-AlmaSoprano-B
Application Alma Soprano Handle Piece Repair And Replacement
|Power per submounts||100W|
|Number of submounts in one stack||12 bars|
|Central wavelength||at 25℃ 808 ± 5nm|
|Typical Operation Current||90A|
|Typical Threshold current||15A|
|Maximum Operating Current||24A|
|Maximum Operating Voltage||24V|
|Fast Axis Divergence (FWHM)||39º|
|Slow Axis Divergence (FWHM)||8º|
|Expected lifetime||10000 hours|
|Typical pitch per bar||1.8 mm|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|
12bars alma laser stack replacement Dimension
There are no reviews yet.