VS-MI-100-12-AlmaSoprano-E-808nm 12bars Alma laser stack
Product Code: VS-MI-100-12-AlmaSoprano-E-808nm
Availability: In Stock
This VS-MI-100-12-AlmaSoprano-E is an Alma Soprano 12-bar laser stack with a black side plate & right-angled electrode.
It is used for Alma Soprano handle piece repairing and replacement.
Oriental-laser focused on diode laser bar assembling for more than 16 years and we have seen various alma soprano diode laser stacks. Also, we can use the same Jenoptik bar like Alma used to recondition the power of the laser soprano power.
Alma soprano ice handle Repairing procedure
>Choice A: Send the old broken handpiece to our factory. We will do the process of checking, repairing, cleaning and etc. Then send the fresh new handpiece back to you.
>Choice B: if you have basic assembly capacity, you can send the diode laser stack only to our factory. We remove the old bar and mount the new bar to recover the laser power of the stack.
>Choice C: Measure your stack and confirm the water pipe hole and screw hole position. Then press the button below to buy a ready-made stack that matches your old handpiece.
Repair Alma soprano ice handle Advantages
>Save Money. Replace the broken diode stack only. Low cost and convenient
>Long lifetime. we use diode laser bars from Germany. The newest Austin mounting technology will be applied when repairing. After repair. The handpiece will be a new one, lifetime is much longer than before.
>All technical support available. We majored in diode laser mounting, software, and spare parts for over 16 years. All the technologies about laser hair removal machines are open to you.
Alma Soprano ice 12bar laser stack Specification
Model No VS-MI-100-12-AlmaSoprano-E
Application Alma Soprano ice Handle Piece Repair And Replacement
|Power per submounts||100W|
|Number of submounts in one stack||12 bars|
|Central wavelength||at 25℃ 808 ± 5nm|
|Typical Operation Current||90A|
|Typical Threshold Current||15A|
|Maximum Operating Voltage||24V|
|Fast Axis Divergence (FWHM)||39º|
|Slow Axis Divergence (FWHM)||8º|
|Expected lifetime||10000 hours|
|Typical pitch per bar||1.8 mm|
|Flow rate||0.3L/min ±10% / Bar|
|Water Quality||Deionized 2-10 µs/cm, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination with copper would form galvanic elements|
Laser stack Dimension
There are no reviews yet.